The Philips Heartstart AED HS1 Defbirillator is designed keeping in mind an ordinary person (lay responder) so he/she can use it in an extraordinary moment. It is easy to set-up and extremely easy to use with automatic Life Guidance features like voice prompts and CPR coaching to help guide one through the treatment of sudden cardiac arrest.
A clean and simple interface that is easy to use and understand. The intelligent voice instructions guides the responder through the entire process starting from placing the pads, CPR and delivering a shock. The intelligent sensors with the help of SMART pads can detect the step one is on and can adjust as per the level of expertise of the responder so that one doesn’t fee lost! If needed it goes back to the level the responder is on, if needed repeat the instructions or even rephrase the instructions.
The defibrillator automatically detects through the SMART pads whether a shock is necessary or a CPR is enough. Where Child/Infant SMART Pads are installed it automatically adjust the energy levels for electric shock appropriate for a child and modify the CPR coaching instructions as per a child.
9 volt DC, 4.2 Ah, composed of disposable long-life lithium manganese dioxide primary cells
Minimum 200 shocks or 4 hours of operating time (EN60601-2-4:2003)
|Battery Install-by Date||
Battery is labeled with an install-by date of at least five years from date of manufacture
|Battery Standby Life||
Four years typical when battery is installed by the install-by date. (Will power the AED in standby state within the specified Standby Temperature range, assuming one battery insertion test and no defibrillation uses)
SMART Truncated exponential biphasic. Waveform parameters adjusted as a function of patient impedance
|Defibrillation peak current - Adult||
32 A (150 J nominal) into a 50 ohm load
|Defibrillation peak current - Pediatric||
19 A (50 J nominal) into a 50 ohm load (using optional infant/child SMART Pads Cartridge)
|Shock-to-Shock Cycle Time||
Typically less than 20 seconds between shocks in a series
Able to deliver a shock after the end of a CPR interval, typically in eight seconds
Detailed voice messages guide responder through use of the defibrillator
Instructions for adult and infant/child CPR available at user selection
Via adhesive pads placed on patient's bare skin as illustrated on pads
Green SMART Pads cartridge handle, green On/Off button, blue i-button, orange Shock button
Ready light, blue i-button, caution light
Philips HS1 Defibrillator Semi Automatic, one lithium battery, one adult SMART pads cartridge, quick reference guide, quick set-up guide and owner’s manual.
8 Years Manufacturer Warranty